Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||8 February 2012|
|PDF File Size:||14.1 Mb|
|ePub File Size:||1.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remember me Forgot password? If the electrocardiograph is not turned off, there is a failure in the electrocardiograph. A separating transformer is a transformer with one or more input winding s separated from the output winding s by at least basic insulation [IEC ].
The electrocardiograph does not print Excessive noise during printing causes the electrocardiograph does not anything or recording result is faint.
Check that the recorder cover is firmly closed. Additional protective earth or separating transformer For B: If you leave electrolyte cream cardioCream on the electrode surface or leave the metal part wet, it may cause inaccurate ECG recording due to high skin electrode contact impedance.
Reattach the electrode and connect the electrode lead. Date display format Select the date display format.
Manual Cardiofax – PDF Free Download
Use only Nihon Kohden approved products with this device. Select the next lead group. There is not enough space to save new ECG data file in the memory. To cancel while copying files, press the Cancel function key. Check the electrode attachment of the indicated position.
decreto legislativo pdf – PDF Files
The characters are toggled. There are two automatic recording modes, real time and review recording mode. When the thermal head is dirty, the recording paper may skew.
Enter the hospital name. Keep this manual near the instrument or in the reach of the operator and refer to it whenever the operation is unclear. Failure to follow this instruction may cause the electrocardiograph to fall over or cause battery liquid leakage. NOTE Formatting deletes all data in the media. When using the output signal from the electrocardiograph for the synchronization signal on other instrument, always consider this time onp.
Copyright Notice The entire contents of this manual are copyrighted by Nihon Kohden.
Contact your Nihon Kohden representative. You can change settings without turning the power off. Steps 5 and 6 are not necessary for some items. Message and key function display area Displays the functions of the function keys and screen message.
Manual Cardiofax 1150
QTc interval is a corrected value based on a heart rate of 60 bpm. Otherwise, the SD memory card and electrocardiograph may get damaged.
The Main Menu screen is displayed. Do not connect unspecified electrical instruments to a multiple portable socket outlet in the system. The AC power lamp and battery charge lamp light when the electrocardiograph battery is charging. If the cause legidlativo the trouble is freezing of the software by static electricity, the electrocardiograph turns off.
Also, noise may superimpose on the ECG waveforms and measurement data may be incorrect.
When managing examination data that is only in electronic file, make sure to enter the patient information.