Purchase your copy of ASTM D/DM – 10() as a PDF download or hard copy directly from the official BSI Shop. All BSI British. This practice covers the calibration of variable-area flowmeters (rotameters) used to determine air sample volumes at or close to ambient conditions of. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||12 September 2018|
|PDF File Size:||13.45 Mb|
|ePub File Size:||8.92 Mb|
|Price:||Free* [*Free Regsitration Required]|
Standards Subscriptions from ANSI provides a money-saving, multi-user solution for accessing standards. Your Alert Profile lists the documents that will be monitored. Last previous aastm D — 73 1. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service e-mail ; or through the ASTM website.
ASTM D3195/D3195M – 10(2015)
Terminology See Test Method E Already Subscribed to this document. Use Methods D A number in parentheses indicates the year of last reapproval. Subscription pricing is determined by: Please first verify your email before subscribing to alerts. This conforms to diaphragm type pump of adequate capacity and a ballast most of the ASTM methods for atmospheric sampling and volume or restrictor f3195 eliminate pulsations.
See bility of regulatory limitations prior to use. Combining values from the two systems may result in non-conformance with the standard. Procedure Using Wet Test Meter 1 7. We have no amendments or corrections for this standard.
Note the manometer reading when room air is used for calibrating gas.
Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters. This standard is also available to be included in Standards Subscriptions.
Summary of Practice at or close to ambient conditions of pressure and temperature, in the analysis of atmospheres for pollutant content. Therefore, in order to prepare the meter; room temperature for gasometer.
It is the responsibility x3195 the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
If the document is revised or amended, you will be notified by email. A superscript epsilon indicates an editorial change since the last revision or reapproval. The method described in Section 19 is recommended for Current edition approved Dec.
Procedure Using Gasometer room temperature, barometric pressure in accordance with Test 8. Since viscosity is propor- added by the wet test meter or gasometer, assuming saturation, tional to the square root of absolute temperature independent by operating with the s3195 The factor for this correction is: Add to Alert PDF.
Proceed to Checkout Continue Shopping. Use average values and meter d1395 temperature for each meter reading. Scope ture scales are to be used when substituting values into the 1.
It is the 4.
If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below. Please first log in with a verified email before subscribing to alerts.
ASTM D/DM – Standard Practice for Rotameter Calibration
You can download and open this file to your own computer but DRM prevents opening this file on another computer, including a networked server. This asm be done logically in several steps. Repeat, going from high to low.
Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend. You may delete a document from your Alert Profile at any time. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other.
Higher accuracy is possible with the gasom- eous Fuel Samples2 eter. Need more than one copy?